Flow Meets Maximum Versatility
The Arrowg+ard Blue® MAC is designed to meet the needs of a variety of patient types — from trauma to cardiac. The catheter combines the access of a sheath introducer with the high-flow lumens of a central line. The MAC's hemostasis valve allows for easy access for additional devices, such as a thermodilution catheter or Arrow® MAC Companion Catheter adding up to three additional lumens. The 9 French distal lumen accommodates large fluid volumes when needed, while the 12-gauge proximal lumen provides a separate access for incompatible fluid administration.
|1.2 - 6.3||1.6||6.3||3-6||12||29|
Note: This table compares the highest flow lumen per vascular access device. PICC, CVC, Sheath Introducer and MAC flow rates are gravity saline flow rates. IV and Intraosseous flow rates are based off pressure bag flow rates, and they are provided by another manufacturer.
|Distal (9 Fr.) No Catheter||29,920|
|Distal (9 Fr.) with 8 Fr. Catheter||12,180|
|Distal (9 Fr.) with 7 Fr. Catheter||19,810|
|Proximal (12 Ga.) No Catheter||9,340|
Based on 100 cm head height gravity saline flow rates. Flow rates would be higher with a rapid infuser or pressure bag.
The Arrow® MAC uses proprietary chlorhexidine-based antimicrobial protection designed to protect your patients from the known pathogens of Catheter-Related Bloodstream Infections (CRBSIs).
Arrow® ErgoPack® Complete systems put necessary components in an intuitive configuration to help optimize your workflow, comply with CDC, SHEA, INS, and OSHA guidelines3-6, and fight against central line-associated bloodstream infections(CLABSI).
Check out what’s new:
Arrow GlideWheel™ Advancer
Provides tactile feedback + finer control
Is kink-resistant compared to stainless steel
To support guidelines for ultrasound insertion
Pre-filled sterile saline syringe
One pre-filled sterile saline syringe per lumen now included inside the kit
Extra ChloraPrep® Skin Prep
Extra applicator now included in each kit
Pressure Injectable VANTEX Central Venous Catheters Indications For Use. Williamston, MI: Centurion Medical Products; 4.
Gupta N, Weber H, Moss S, Gaire-Patel K. Are antibiotic resistant “super bugs” a real challenge to antimicrobial central venous catheter performance? AVA 2014.
O'Grady NP, Alexander M, Burns LA, et al. Guidelines for the Prevention of Intravascular Catheter-Related Infections, 2011 (Revised 2017). 2. Atlanta, GA: Centers for Disease Control and Prevention; 2017.
Marschall J, Mermel LA, Fakih M, et al. Strategies to Prevent Central Line–Associated Bloodstream Infections in Acute Care Hospitals: 2014 Update. Infection Control and Hospital Epidemiology. 2014 July; 35(7): 753-71.
Gorski L, Hadaway L, Hagle ME, McGoldrick M, et al. Infusion Therapy Standards of Practice. Journal of Infusion Nursing. 2016; Jan 39(1S).
Occupational Safety & Health Administration Regulations (Standards – 29 CFR). Part 1910.1030: Bloodborne pathogens. Occupational Safety & Health Administration Web site. https://www.osha.gov/laws-regs/regulations/standardnumber/1910/1910.1030. Accessed on February 12, 2020.